- No products in the cart.
Read through our frequently asked questions below
|After completing dynamic wand wave it is possible for the Set Origin operation to fail. This is rare, but for those who experience the problem, it will be consistent.||2.10.3||The only way to get it to work again is to close Nexus down and reopen.||Solved
Resolved in Nexus 2.11.0
|Nexus performance issue when there was a large number of log messages.||2.10.1||Confirmed log workaround in Nexus 2.10.1:
Note that this will disable all log messages, both in the app and in the log file.
If you want to keep the log messages:
Resolved in Nexus 2.10.2
|GetDeviceChannel returns force plate outputs as an action force in a z-down coordinate system, compared to a reaction in a z-up system as stored in the C3D file.||2.10.1||Perform the conversion after requesting the value: rotate the force and moment outputs by 180 degrees about the y-axis, and then negate all components: this is equivalent to simply negating the y-component. This applies to both force and moment outputs.||Scheduled to be resolved|
|PC blue screen when connected digitally to Kistler force plates due to Kistler digital plug-in||2.6.1||With Nexus open: